Payment & Shipping Terms:
50Hz Air Flow Test Equipment 1Kw Air Cleaner 30M³ Chamber Equipped With Glass
Model: JHQ-1 Environment test chamber of air cleaner
Overview: According to relevant requirements stipulated in GB/T 18801-2015 Air Cleaner, this
device is applied to the solid pollutant removal testing(natural attenuation test of solid pollutants,
total attenuation test of solid pollutants, calculation of clean air delivery rate of solid pollutant
removal of air cleaner), gaseous pollutant removal test (natural attenuation test of gaseous
pollutant, total attenuation test of gaseous pollutant, calculation of clean air delivery rate of solid
pollutant removal of air cleaner ), service test of air cleaner and other test.
1. Dimension: 3400 ×3500 ×2500 (3400 ×3500 ×2500), allowable tolerance ±0.5m³
2. Structure: Warehouse board or frame is equipped with glass.
3. Observation window: 1000*700.
4. Door: 2000*800.
5. Sealing material: Use silicon rubber bar and glass sealant with low volatility.
6. Mixing fan: Diameter 1400mm, metal vane, degree of mixing of chamber is greater than 80%.
7. Circulating fan: (500-700)m3/h, diameter 20cm, installation location: 1.5m above the ground,
0.4m away from rear wall.
8. Sampling hole : 3+3 pollutant filling and sampling port: Located at the front wall and the right
side wall respectively; ball valve control, with flexible hose.
9. Lighting: 2 sets of cleaning luminaries with dual tubes.
10. Cleaning system (internal cleaning system): Primary efficiency filter screen, intermediate
efficiency filter screen, high efficiency filter screen, activated carbon filter screen (activated
carbon filter screen of formaldehyde testing chamber shall be thickened properly).
11. Temperature and humidity sensor: Humidity: 30-90%RH±3%RH; temperature: 10-50℃
12. Air circulation flow, refrigerating capacity, heating capacity, humidifying capacity and other
parameters shall be determined by yourself according to operating conditions said above.
However, the time taken from normal temperature to steady working condition shall not be
greater than 20min, and the standard temperature is (25±2) ℃and the relevant humidity is
(50±10)% during test of testing chamber.
13. Power supply of prototype: Single-phase 50-300V, 50Hz, capacity 1kW.
14. Control parameters: Under automatic testing state, all working processes are in accordance